Monitoring and evaluation of lymphatic filariasis interventions: an improved PCR-based pool screening method for high throughput Wuchereria bancrofti detection using dried blood spots
© Plichart and Lemoine; licensee BioMed Central Ltd. 2013
Received: 9 December 2012
Accepted: 14 April 2013
Published: 18 April 2013
Effective diagnostic tools are necessary to monitor and evaluate interruption of Lymphatic Filariasis (LF) transmission. Accurate detection of Wuchereria bancrofti (Wb) microfilaria (mf) is essential to measure the impact of community treatment programmes. PCR-based assays are specific, highly sensitive tools allowing the detection of Wuchereria bancrofti DNA in human blood samples. However, current protocols describing the pool screening approach, use samples of less than 60 μl of blood, which limits the sensitivity of the pool-screen PCR assay. The purpose of this study was to improve the pool-screen PCR protocol to enhance its sensitivity and usefulness for population scale studies.
DNA extractions were performed with the DNeasy kit, the PCR with the Wb LDR primers and the SYBR-Green dye. Improvements of our pool-screen real-time PCR (qPCR) assay allowed the detection of as little as one Wb microfilaria diluted in a pool of at least 12 blood samples of 60 μl each. Using this assay, mf burdens can be predicted using a standard curve derived from mf spiked dried blood samples. The sensitivity achieved is equivalent to the detection of a single LF positive individual carrying a mf burden as low as 18 mf/ml, in a pool of blood samples from at least 12 individuals.
Due to its sensitivity, rapidity and cost-effectiveness, we suggest this qPCR pool-screening assay could be used as a diagnostic tool for population- scale filariasis elimination monitoring and evaluation.
The main goal of the Global Program to Eliminate Lymphatic Filariasis (GPELF) is to interrupt disease transmission by reducing microfilaremia prevalence in blood through mass drug administration (MDA) . Diagnostic tools are required to assess the status of LF in countries that are now in the post-MDA surveillance phase or are still implementing preventive chemotherapy. The rapid immunochromatographic test (ICT), detecting filarial antigen, is the selected tool for deciding when to stop MDA  in areas where levels of infection have been reduced to a point where transmission is no longer sustainable . However, because antigenemia decreases more slowly than microfilaremia [4–6], the detection of mf in human blood populations of sentinel and spot-check sites remains an essential complementary test for assessing the impact of MDA [3, 7–9]. Several methods are available for mf testing: the counting chamber method , the microscopic examination of capillary blood films (60-μl thick) [11–14] or membrane filtration from one ml of venous blood . Because these techniques can only be realized on individual samples, they are labour intensive and not adapted for monitoring residual mf prevalence rates in the community. As mf prevalence decreases through MDA, the pool screening approach becomes necessary and more cost effective . PCR assays have been developed [16, 17] that are highly sensitive and specific for the detection of Wb DNA in individual human blood samples [18–20] as well as in mosquito vectors [21–25]. Conventional PCR and qPCR-based pool-screening methods, using pools of up to 10 blood samples of 10 μl to 30 μl each, have already been described [2, 26–28]. However, they are less sensitive than the thick blood film method usually used , due to the low volume of blood analysed per patient. Thus, improving the pool-screening sensitivity of the qPCR assay by increasing the volume of blood tested would allow the broader implementation of this diagnostic tool.
The purpose of this study was to improve the DNA extraction protocol and qPCR assay to achieve the detection of a single mf in a pool of at least twelve 60-μl blood samples.
Human blood samples
The assay was performed using capillary blood dried on filter paper as capillary blood is used to test microfilaremia in most countries endemic for LF. All procedures were carried out in the laboratory. All samples were acquired under protocols approved by the French Polynesia Ethics Committee with written informed consent obtained from all subjects. For each subject, 60 μl of blood was loaded on 6 spots of a filter paper disk (TropBio, Townsville, Australia), dried and stored at −20°C until use. Fifty-eight samples negative for filarial antigen (ICT and Og4C3) and for microfilariae (PCR and blood smear) were used for the assays. Samples containing one mf were obtained by very carefully spiking a single mf purified from the blood of a microfilaremic individual, onto the filter paper disk of a non-infected subject as identified above. Samples containing 3 mf and 10 mf were obtained similarly. Positive samples of approximately 100 mf were prepared with 60 μl of blood from an infected person whose microfilaremia was estimated at 1548 mf/ml by the filtration method. The six blood filter spots of each sample were processed.
DNA was extracted from the blood filter spots using the QIAGEN DNeasy kit (Qiagen, Hilden, Germany; Cat N° 69504) following the manufacturer’s instructions and increasing the volumes of reagents [Laney SJ, unpublished data]. Briefly, blood spots of a subject were placed in a 2 ml microtube, covered with 270 μl of ATL buffer, incubated at 85°C for ten minutes, then at 56°C for one hour after addition of 30 μl of Proteinase K. Addition of 300 μl of Al buffer to the digested suspension brought up the lysate volume to a total of 600 μl. From this step, we used two different protocols. First, the lysate was either heat-treated (100°C) or not, prior to the DNA purification. The aim of the heat treatment was to denature the genomic DNA to make the DNA target sequence more accessible to the primers. Second, individual lysates (total or fraction) or pooled lysates were processed in the subsequent DNA purification step. A volume of ethanol equivalent to half the volume of lysate sample was added before loading the mix onto a DNeasy spin column. Depending on the volume loaded, additional centrifugations were performed to pass all the solution through the column. After washing steps (twice with AW1 buffer, once with AW2 buffer), purified DNA was eluted in 200 μl of AE buffer.
qPCR assays were performed on the Bio-Rad real-time thermal cycler CFX96, using the Wb-LDR primers  and the SYBR Green fluorescence dye with melting curve analysis. The target sequence size is 90 bp long. Each reaction contained 12.5 μl of IQ SYBRGreen supermix 2× (Biorad), 0.75 μl of 10 pmol μl-1 of forward and reverse primers and 5 μl of DNA template in a total volume of 25 μl. Thermal cycling conditions for PCR were 95°C for 3 minutes followed by 40 cycles of 10 s at 95°C and 30s at 60°C. The melt curve analysis was performed by reducing the temperature to 55°C for 1 minute and then raising the temperature by increments of 0.5°C every 10 s up to 95°C. The results were analysed with the CFX manager software to calculate the Ct value corresponding to the number of reaction cycles necessary to detect a signal above baseline.
Assays and results
Before performing the assays with pool samples, the impact of the heat treatment prior to the DNA purification was assessed for each individual sample. To that end, lysates were heated at 100°C for 5 minutes then cooled immediately on ice. DNA purification and qPCR were then performed as described above. The lysates from either 0 mf, 1 mf or 100 mf blood samples were tested in duplicate.
qPCR results depending on lysate volume processed and on microfilaria (mf) burden
mf per sample
Volume of lysate (μl)
Ct values depending on lysate volume and lysate number per pool
Sample volume (μl)
The experiment was repeated using only 50 μl of lysate per sample to save time by reducing the number of centrifugations necessary during the DNA purification step. We have processed three pools of 10 lysates and one pool of 12 lysates, and obtained mean Ct values of 33.3 and 33.6 respectively (Table 2). The results showed that as little as 50 μl of lysate could be used to constitute the pools without loss of sensitivity.
Using the Ct values obtained with all low positive and negative samples tested, an arbitrary threshold was set at a Ct value of 36 (threshold baseline at 500) to exclude false positive results. All results with a Ct value below 36 and a melt peak temperature of 76.5°C were therefore considered positive.
One of the principal measures of success of the GPELF is the decrease in microfilaremia prevalence and in mf load, a sine qua non condition to LF elimination. Our improvement of the PCR-based pool screening method allows for a highly sensitive screening tool relevant for LF control. Although this assay currently requires real-time PCR equipment, it may ultimately be adapted for use in resource-poor endemic areas . The present assay may stimulate the development and support of much-needed national and regional reference laboratories to suit the various LF epidemiological situations and populations at risk.
We would like to thank Tokahi Rotillon for her help and technical assistance. We are thankful to Dr. Bossin for his critical contribution to this manuscript. We gratefully acknowledge Dr Sandra J. Laney for her critical reading of the manuscript.
- Ottesen EA, Hooper PJ, Bradley M, Biswas G: The global programme to eliminate lymphatic filariasis: health impact after 8 years. PLoS Negl Trop Dis. 2008, 2 (10): e317-10.1371/journal.pntd.0000317.PubMed CentralView ArticlePubMedGoogle Scholar
- Gass K, de Rochars MV B, Boakye D, Bradley M, Fischer PU, Gyapong J, Itoh M, Ituaso-Conway N, Joseph H, Kyelem D: A multicenter evaluation of diagnostic tools to define endpoints for programs to eliminate bancroftian filariasis. PLoS Negl Trop Dis. 2012, 6 (1): e1479-10.1371/journal.pntd.0001479.PubMed CentralView ArticlePubMedGoogle Scholar
- WHO: Global Programme to Eliminate Lymphatic Filariasis. Monitoring and epidemiological assessment of mass drug administration: a manual for national elimination programmes. 2011, Geneva: World Health Organization, WHO/HTM/NTD/PCT/2011.4Google Scholar
- Schuetz A, Addiss DG, Eberhard ML, Lammie PJ: Evaluation of the whole blood filariasis ICT test for short-term monitoring after antifilarial treatment. Am J Trop Med Hyg. 2000, 62 (4): 502-503.PubMedGoogle Scholar
- Tisch DJ, Bockarie MJ, Dimber Z, Kiniboro B, Tarongka N, Hazlett FE, Kastens W, Alpers MP, Kazura JW: Mass drug administration trial to eliminate lymphatic filariasis in Papua New Guinea: changes in microfilaremia, filarial antigen, and Bm14 antibody after cessation. Am J Trop Med Hyg. 2008, 78 (2): 289-293.PubMed CentralPubMedGoogle Scholar
- Simonsen PE, Pedersen EM, Rwegoshora RT, Malecela MN, Derua YA, Magesa SM: Lymphatic filariasis control in Tanzania: effect of repeated mass drug administration with ivermectin and albendazole on infection and transmission. PLoS Negl Trop Dis. 2010, 4 (6): e696-10.1371/journal.pntd.0000696.PubMed CentralView ArticlePubMedGoogle Scholar
- Molyneux DH: Filaria control and elimination: diagnostic, monitoring and surveillance needs. Trans R Soc Trop Med Hyg. 2009, 103 (4): 338-341. 10.1016/j.trstmh.2008.12.016.View ArticlePubMedGoogle Scholar
- Weil GJ, Ramzy RM: Diagnostic tools for filariasis elimination programs. Trends Parasitol. 2007, 23 (2): 78-82. 10.1016/j.pt.2006.12.001.View ArticlePubMedGoogle Scholar
- Ramzy RM, El Setouhy M, Helmy H, Ahmed ES, Abd Elaziz KM, Farid HA, Shannon WD, Weil GJ: Effect of yearly mass drug administration with diethylcarbamazine and albendazole on bancroftian filariasis in Egypt: a comprehensive assessment. Lancet. 2006, 367 (9515): 992-999. 10.1016/S0140-6736(06)68426-2.View ArticlePubMedGoogle Scholar
- Wamae CN, Njenga SM, Ngugi BM, Mbui J, Njaanake HK: Evaluation of effectiveness of diethylcarbamazine/albendazole combination in reduction of Wuchereria bancrofti infection using multiple infection parameters. Acta Trop. 2011, 120 (Suppl 1): S33-38.View ArticlePubMedGoogle Scholar
- Boyd A, Won KY, McClintock SK, Donovan CV, Laney SJ, Williams SA, Pilotte N, Streit TG, de Rochars MV B, Lammie PJ: A community-based study of factors associated with continuing transmission of lymphatic filariasis in Leogane, Haiti. PLoS Negl Trop Dis. 2010, 4 (3): e640-10.1371/journal.pntd.0000640.PubMed CentralView ArticlePubMedGoogle Scholar
- Mathieu E, Dorkenoo A, Otogbe FK, Budge PJ, Sodahlon YK: A laboratory-based surveillance system for Wuchereria bancrofti in Togo: a practical model for resource-poor settings. Am J Trop Med Hyg. 2011, 84 (6): 988-993. 10.4269/ajtmh.2011.10-0610.PubMed CentralView ArticlePubMedGoogle Scholar
- Iboh CI, Okon OE, Opara KN, Asor JE, Etim SE: Lymphatic filariasis among the Yakurr people of Cross River State. Nigeria. Parasit Vectors. 2012, 5: 203-10.1186/1756-3305-5-203.View ArticlePubMedGoogle Scholar
- Koroma JB, Bangura MM, Hodges MH, Bah MS, Zhang Y, Bockarie MJ: Lymphatic filariasis mapping by immunochromatographic test cards and baseline microfilaria survey prior to mass drug administration in Sierra Leone. Parasit Vectors. 2012, 5: 10-10.1186/1756-3305-5-10.PubMed CentralView ArticlePubMedGoogle Scholar
- Esterre P, Plichart C, Sechan Y, Nguyen NL: The impact of 34 years of massive DEC chemotherapy on Wuchereria bancrofti infection and transmission: the Maupiti cohort. Tropical medicine & international health: TM & IH. 2001, 6 (3): 190-195. 10.1046/j.1365-3156.2001.00682.x.View ArticleGoogle Scholar
- Zhong M, McCarthy J, Bierwert L, Lizotte-Waniewski M, Chanteau S, Nutman TB, Ottesen EA, Williams SA: A polymerase chain reaction assay for detection of the parasite Wuchereria bancrofti in human blood samples. Am J Trop Med Hyg. 1996, 54 (4): 357-363.PubMedGoogle Scholar
- Rao RU, Atkinson LJ, Ramzy RM, Helmy H, Farid HA, Bockarie MJ, Susapu M, Laney SJ, Williams SA, Weil GJ: A real-time PCR-based assay for detection of Wuchereria bancrofti DNA in blood and mosquitoes. Am J Trop Med Hyg. 2006, 74 (5): 826-832.PubMed CentralPubMedGoogle Scholar
- Williams SA, Nicolas L, Lizotte-Waniewski M, Plichart C, Luquiaud P, Nguyen LN, Moulia-Pelat JP: A polymerase chain reaction assay for the detection of Wuchereria bancrofti in blood samples from French Polynesia. Trans R Soc Trop Med Hyg. 1996, 90 (4): 384-387. 10.1016/S0035-9203(96)90515-5.View ArticlePubMedGoogle Scholar
- McCarthy JS, Zhong M, Gopinath R, Ottesen EA, Williams SA, Nutman TB: Evaluation of a polymerase chain reaction-based assay for diagnosis of Wuchereria bancrofti infection. J Infect Dis. 1996, 173 (6): 1510-1514. 10.1093/infdis/173.6.1510.View ArticlePubMedGoogle Scholar
- Fischer P, Boakye D, Hamburger J: Polymerase chain reaction-based detection of lymphatic filariasis. Medical microbiology and immunology. 2003, 192 (1): 3-7.PubMedGoogle Scholar
- Williams SA, Laney SJ, Bierwert LA, Saunders LJ, Boakye DA, Fischer P, Goodman D, Helmy H, Hoti SL, Vasuki V: Development and standardization of a rapid, PCR-based method for the detection of Wuchereria bancrofti in mosquitoes, for xenomonitoring the human prevalence of bancroftian filariasis. Ann Trop Med Parasitol. 2002, 96 (Suppl 2): S41-46.View ArticlePubMedGoogle Scholar
- Plichart C, Sechan Y, Davies N, Legrand AM: PCR and dissection as tools to monitor filarial infection of Aedes polynesiensis mosquitoes in French Polynesia. Filaria journal. 2006, 5: 2-10.1186/1475-2883-5-2.PubMed CentralView ArticlePubMedGoogle Scholar
- Plichart C, Laney SJ, Sechan Y, Davies N, Legrand AM: Correction PCR and dissection as tools to monitor filarial infection of Aedes polynesiensis mosquitoes in French Polynesia. Filaria journal. 2007, 6: 5-10.1186/1475-2883-6-5.PubMed CentralView ArticleGoogle Scholar
- Boakye DA, Baidoo HA, Glah E, Brown C, Appawu M, Wilson MD: Monitoring lymphatic filariasis interventions: Adult mosquito sampling, and improved PCR - based pool screening method for Wuchereria bancrofti infection in Anopheles mosquitoes. Filaria journal. 2007, 6: 13-10.1186/1475-2883-6-13.PubMed CentralView ArticlePubMedGoogle Scholar
- Farid HA, Morsy ZS, Helmy H, Ramzy RM, El Setouhy M, Weil GJ: A critical appraisal of molecular xenomonitoring as a tool for assessing progress toward elimination of Lymphatic Filariasis. Am J Trop Med Hyg. 2007, 77 (4): 593-600.PubMed CentralPubMedGoogle Scholar
- Supali T, Ismid IS, Wibowo H, Djuardi Y, Majawati E, Ginanjar P, Fischer P: Estimation of the prevalence of lymphatic filariasis by a pool screen PCR assay using blood spots collected on filter paper. Trans R Soc Trop Med Hyg. 2006, 100 (8): 753-759. 10.1016/j.trstmh.2005.10.005.View ArticlePubMedGoogle Scholar
- Ramzy RM, Farid HA, Kamal IH, Ibrahim GH, Morsy ZS, Faris R, Weil GJ, Williams SA, Gad AM: A polymerase chain reaction-based assay for detection of Wuchereria bancrofti in human blood and Culex pipiens. Trans R Soc Trop Med Hyg. 1997, 91 (2): 156-160. 10.1016/S0035-9203(97)90205-4.View ArticlePubMedGoogle Scholar
- Fink DL, Kamgno J, Nutman TB: Rapid molecular assays for specific detection and quantitation of Loa loa microfilaremia. PLoS Negl Trop Dis. 2011, 5 (8): e1299-10.1371/journal.pntd.0001299.PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.