- Open Access
Development of a multiplex fluorescence immunological assay for the simultaneous detection of antibodies against Cooperia oncophora, Dictyocaulus viviparus and Fasciola hepatica in cattle
© Karanikola et al. 2015
- Received: 1 May 2015
- Accepted: 1 June 2015
- Published: 19 June 2015
A major constraint for the effective control and management of helminth parasites is the lack of rapid, high-throughput, routine diagnostic tests to assess the health status of individual animals and herds and to identify the parasite species responsible for these helminthoses. The capability of a multiplex platform for the simultaneous detection of three pasture associated parasite species was evaluated and compared to existing ELISAs.
The recombinant antigens 14.2 kDa ES protein for Cooperia oncophora, major sperm protein for Dictyocaulus viviparus and Cathepsin L1 for Fasciola hepatica were recombinantly expressed either in Escherichia coli or Pichia pastoris. Antigens were covalently coupled onto magnetic beads. Optimal concentrations for coupling were determined following the examination of serum samples collected from experimentally mono-infected animals, before and after their infection with the target species. Absence of cross-reactivity was further determined with sera from calves mono-infected with Haemonchus contortus, Ostertagia ostertagi and Trichostrongylus colubriformis. Examination of negative serum samples was characterised by low median fluorescence intensity (MFI).
Establishment of the optimal serum dilution of 1:200 was achieved for all three bead sets. Receiver Operating Characteristic analyses were performed to obtain cut-off MFI values for each parasite separately. Sensitivity and specificity at the chosen cut-off values were close to, or 100 % for all bead sets. Examination of serum samples collected on different days post infection from different animals showed a high reproducibility of the assays. Serum samples were additionally examined with two already established ELISAs, an in-house ELISA using the recombinant MSP as an antigen and a DRG ELISA using Cathepsin L1 for liver fluke. The results between the assays were compared and kappa tests revealed an overall good agreement.
A versatile bead-based assay using fluorescence detection (xMAP® technology) was developed to simultaneously detect antibodies against C. oncophora, D. viviparus and F. hepatica in cattle serum samples. This platform provides rapid, high-throughput results and is highly sensitive and specific in comparison to existing serological as well as coproscopical diagnostic techniques.
- Parasitic gastroenteritis
- Liver fluke
- Multiplex immunoassays
Nematode and trematode infections play an important role for animal welfare and are of great concern for the economy of the global ruminant livestock industry today . Constantly increasing financial costs for anthelminthic prophylaxis and treatment due to the spread of anthelmintic resistant parasite populations, as well as the often overlooked subclinical effects of the helminth infections on animal productivity [2, 3] have led to the need of developing new and sustainable strategies concerning the effective control of helminthoses. An important step towards this end is the development of new, efficient and high-throughput diagnostic techniques. Despite the development of more sensitive coproscopical methods , they often target individual animals and are not suitable for high-throughput diagnosis. Serological methods established so far appear to lack specificity, in particular when non-recombinant antigens are used, and multiple tests have to be performed to detect mixed species infections.
Amongst the helminths responsible for pasture-borne parasitoses the liver fluke, lungworm as well as gastrointestinal (GI) nematodes are the most important for cattle in temperate climate regions. Cooperia oncophora, a parasite of the small intestine of cattle occurs worldwide with high prevalence rates [5, 6]. It is usually associated with Ostertagia ostertagi, parasitising the abomasum, and both contribute to the complex of parasitic gastroenteritis (PGE) . While O. ostertagi is known to be more pathogenic, C. oncophora is generally considered as the dose-limiting species , particularly for the macrocyclic lactone (ML) anthelmintics. Infections with these GI nematodes as well as with the bovine lungworm Dictyocaulus viviparus cause considerable decrease in weight gain in calves during the first grazing  season, can be responsible for significant reduction in milk yield [9, 10] and impair animal welfare . The economic importance of such parasitic diseases has been repeatedly demonstrated , especially where intensive grazing management is practiced, and cost-benefit evaluations have indicated that financial losses due to the presence of GI nematodes can be high [2, 13]. Monitoring data provided by the Dutch Animal Health Service for D. viviparus indicate that the incidences of parasitic bronchitis tended to increase in the Netherlands . In the same study, economic losses of approximately 160 € per cow were calculated. The liver fluke Fasciola hepatica affects large and small ruminants. In cattle, fasciolosis can appear as a chronic and subclinical form and is worldwide considered as one of the most important parasitic diseases causing substantial economic losses, which are estimated to be 2000 million $ (USA) per year in agriculture [15, 16]. Additionally, this parasite has zoonotic potential and environmental contamination through infected animals can be important for human health .
Diagnosis of these parasites is commonly based on coproscopical detection methods such as sedimentation (liver fluke), flotation (GI nematodes) or baermannisation of larvae (lungworm). Since eggs excreted by most GI nematodes are morphologically indistinguishable, species identification can only be achieved following faecal culturing or using molecular techniques [18–20]. The generally high handling costs as well as the necessity to sample several animals led to the increased use of serological methods which can be used for herd health monitoring. Serological diagnosis of F. hepatica has been described in the literature using excretory/secretory (ES) products [21–23], a “f2” antigen (Fasciolosis Verification Test, IDEXX, Hoofddorp, the Netherlands) and a recombinant Cathepsin L1 antigen . The same applies for D. viviparus where the detection of antibodies in serum or milk using ELISAs with either crude ES antigen [24–26] or recombinantly expressed major sperm protein (MSP) [27–29] has been described. For the detection of C. oncophora, ELISA using crude antigen , or a recombinantly expressed 14.2 kDa ES protein  were reported.
All these ELISA only target a single species and in order to cover the spectrum of pasture-borne helminthoses, multiple assays have to be conducted. Recent technical advances offer the advantage of multiplex assays, resulting in higher throughput, increased flexibility, reduced sample volume and lower costs [32–34]. A popular multiplex platform is the bead-based Luminex® xMAP® technology (Luminex Corp., Austin, TX). The basis are different polystyrene beads which are labelled with distinct ratios of two fluorescent dyes (red and near-infrared), leading to more than 100 sets of distinguishable beads, which are also referred to as microspheres in the literature. With each set, different analytes can be measured in parallel in a single assay [35, 36]. Various ligands can be covalently conjugated to the surface of these beads. If antigens are used as ligands, assays equivalent to ELISAs can be developed. Interactions of the target analytes with antibodies is detected using biotinylated secondary antibodies and streptavidin, labelled with the reporter fluorochrome phycoerythrin (SA-PE). Fluorescence detection in the Luminex xMAP liquid suspension array system is achieved by two-laser flow cytometry . This technology is relatively widely used in the field of human medical diagnosis  but only few reports have been published in the field of veterinary diagnosis, particularly regarding serological assays [39–41].
The aim of this study was to develop a new, versatile diagnostic assay for the simultaneous detection of specific antibodies against F. hepatica, D. viviparus and C. oncophora in cattle serum samples. The performance of the Luminex® platform was evaluated through comparison with already established ELISAs using the same or different antigens.
Serum samples and antigens
The standardisation of the assay was achieved by using control sera obtained from parasite naïve animals before (negative control) and after experimental mono-infection with the target parasites D. viviparus, (100 larvae over 5 consecutive days) C. oncophora (30,000–40,000 larvae) and F. hepatica (500 metacercariae). For testing specificity as well as cross reactivity, sera from animals mono-infected with other important GI nematodes, Haemonchus contortus, Trichostrongylus colubriformis and O. ostertagi were used. All animal experiments were conducted in strict accordance with the respective local legislation and the European guideline for animal experiments (2010/63/EU). They were approved by a) the Landesamt für Gesundheit und Soziales, Berlin, Germany under the reference number L 0088/10, b) the Ethical Commission of the Faculty of Veterinary Medicine, Ghent University, Belgium under the reference number EC2009/086 and c) the Swedish Animal Ethics Committee under the permission C4/2.
Additionally serum samples collected in Denmark (n = 39), Switzerland (n = 76) and Poland (n = 367) were used. In Denmark and Switzerland, samples were taken from grazing young cattle on randomly selected farms. The sampling in Poland took place on farms previously identified for a cross-sectional survey using a two-stage sampling approach . On a subset of those farms, 10–15 first season grazing cattle were randomly sampled.
The antigen used for the detection of F. hepatica was a recombinant 37 kDa Cathepsin L1-like protein  provided by ILDANA BIOTECH, UCD, Dublin. It is an active site [Cys26Gly] mutant expressed in the yeast Pichia pastoris. For the detection of antibodies against D. viviparus the recombinant 43 kDa MSP expressed as a glutathione-S-transferase (GST) fusion protein in Escherichia coli BL21 (DE3) cells as previously described by Gozdzik et al.  was used.
Production of recombinant C. oncophora ES 14.2 antigen
The protein used for the detection of antibodies against C. oncophora was a 14.2 kDa ES protein described previously by Poot et al. . A codon optimised (E. coli) version of the open reading frame (ORF) was synthesised in vitro (SynthesisGene®; China). The ORF was amplified using the forward primer (5′-CAC CAA TGA ATA TAC CGA TGC ACT GGC AAA ATG TAC-3′) and reverse primer (5′-TTA TTC CCA ATA CAG ACA CAG AAC TTT CAG TT-3′). PCR products were cloned into the pET151 TOPO expression vector (Life Technologies). A Rosetta gami® (Novagene) E. coli clone containing the pET151/CoES14.2 was cultured at 37 °C until OD600nm reached 0.6. Synthesis of the ES14.2-V5-6 × His protein was induced with 0.5 mM isopropylthio-galactoside (IPTG) at 37 °C for 4 h. The recombinant ES14.2-V5-6 × His protein was purified from inclusion bodies using Protino® Ni-IDA columns (Macherey-Nagel, Germany) according to the manufacturer’s protocol. An additional wash step using a 50 mM concentration of imidazole and 2 % Tween20 was conducted before elution with 250 mM imidazole. Purity of the eluted protein was analysed on 12 % SDS–PAGE, stained with GelCode™ colloidal coomassie stain (ThermoFisher). Western blotting using an anti-V5 antibody (Life Technologies) was carried out to confirm that the target protein was obtained.
Antigen coupling to fluorescent beads
In order to remove sodium azide or imidazole, D. viviparus and C. oncophora recombinant antigens were purified by gel filtration using Micro Bio-Spin 6 chromatography columns (Bio-Rad, Germany) according to the manufacturer’s protocol. Concentrations of all antigens were determined using the CB-X™ Assay (G-Biosciences, USA).
D. viviparus, C. oncophora and F. hepatica antigens were conjugated on the surface of carboxylated magnetic beads (Bio-Plex Pro™ Magnetic COOH Beads – 1.25 × 107 beads/ml, Bio-Rad) using the fluorescence regions 026, 062 and 065, respectively. Coupling reactions were performed using the Amine Coupling Kit® (Bio-Rad), following a two-step carbodiimide reaction protocol provided by the manufacturer. The stock suspension of uncoupled beads was vortexed at high speed for 30 s followed by sonication for 15 s in order to disperse bead aggregates. A 100 μl aliquot of monodisperse COOH beads (1.25 × 106) was transferred to one Bio-Plex coupling reaction tube and was placed into the magnetic separator for 30 to 60 s before removal of the supernatant. The beads were washed once in 100 μl bead wash buffer, followed by re-suspension in 80 μl bead activation buffer. Then 10 μl of 50 mg/ml N-Hydroxysulfosuccinimide sodium salt (S-NHS) (Sigma-Aldrich, Germany) and 10 μl of 50 mg/ml N-(3-Dimethylaminopropyl)-N’-ethylcarbodiimide hydrochloride (EDAC) (Sigma-Aldrich), which were prepared in bead activation buffer immediately prior to their use, were added. The reaction tube was mixed gently, covered with aluminium foil and then gently agitated on a shaker for 20 min at room temperature (RT). PBS (pH 7.4, 150 μl) was added twice, always followed by vigorous vortexing. The recombinant protein was added and total volume was brought to 500 μl with PBS (final antigen concentration 5–12 μg/500 μl). Incubation was performed on a shaker at high speed (600–700 rpm) at RT for 2 h. In order to achieve higher coupling yields, some alterations of the initial protocol were made. It was observed that beads incubated at medium speed (500 rpm) as recommended had a tendency to precipitate at the bottom of the coupling reaction tube. Therefore, speed was slightly increased and the coupling reaction tube was vortexed once at high speed after 1 h to prevent precipitation.
Initially, different concentrations were used for each of the three antigens and were separately tested in order to determine the optimum antigen concentration. The amounts of the conjugated protein were the following for each bead-set: for C. oncophora 7 μg, 5 μg, 3.5 μg, 2.5 μg, 1.75 μg, 1.25 μg, 0.9 μg and 0.45 μg, for D. viviparus 5 μg, 2.5 μg, 1.25 μg, 0.66 μg and 0.45 μg, and for F. hepatica 5 μg, 2.5 μg and 1.25 μg. The coupled beads were placed into a magnetic separator for 1 min and after removal of the supernatant they were washed with 500 μl of PBS. The coupled beads were then re-suspended in 250 μl of blocking buffer and gently agitated at RT in the darkness for 30 min. Finally, the beads were washed with 500 μl of storage buffer, re-suspended in 150 μl storage buffer and stored at 4 °C in the dark. Beads were stored on the recommended conditions and always used within 4 months after coupling since decreased performance was observed thereafter.
Luminex multiplex assay
The assays were conducted in 96-well polystyrene, round-bottom microplates (Greiner Bio-One). The three bead-sets were initially tested in singleplex assays including negative controls as well as the respective positive control sera. These were used in a two-fold eight serial dilution series in PBS/Tween20 (0.05 %, pH 7.4) in order to identify the optimal sample dilution. Cross-reactivity was assessed by running the assay with sera from calves infected with non-target species. Then, the three bead-sets were combined in a bead-mix and a multiplex assay was performed.
Prior to each examination, beads were re-suspended by vortexing and sonication for approximately 20 s three times to avoid high numbers of aggregated beads. A 50 μl aliquot of the working beadmixture (concentration 100 beads/μl) was transferred into the wells, followed by the addition of 50 μl of diluted sera. The plate was incubated on a plate shaker (800 rpm) in the dark at RT for 60 min. The plate was then placed into the magnetic separator and left for separation for 60 s. The supernatant was carefully removed from each well by manual inversion. Beads were washed 5 times by adding 100 μl PBS/Tween20 into each well to ensure absence of any undesirable or non-specifically bound antibodies. The plate was then removed from the magnetic separator and 100 μl of a biotinylated secondary antibody (goat anti-bovine IgG, Dianova, Germany) diluted 1:1000 in PBS/Tween20 were added to each well. Incubation was again conducted in darkness and at RT on a plate shaker (800 rpm) for 30 min before beads were washed as described above. Finally, 100 μl of streptavidin-phycoerythrin (SA-PE, Millipore) at 2 μg/ml, diluted in assay buffer, were added to each well. The plate was placed on the shaker, covered with aluminium foil and again incubated at RT on a plate shaker (800 rpm) for 30 min. The supernatant was carefully removed after magnetic separation of the beads by manual inversion and washing was performed as previously described. Assay buffer (100 μl) was added into each well and the plate was placed onto a plate shaker for approximately 15 s in order to achieve gentle agitation of the beads.
The beads were analysed using the Bio-Plex 200 instrument following the manufacturer’s instructions. A minimum of 100 events (beads) per well was read for every bead-set. All samples were analysed in duplicates in each run. To investigate reproducibility of the assays, several (between three and six) runs were performed using the sera from the same animals.
ELISA used for comparison
Result obtained using the Luminex® assay were compared to existing ELISAs. D. viviparus antibodies were detected using either an in-house ELISA for lungworm based on the recombinant MSP antigen as described in von Holtum et al.  or using the modification of this ELISA as described by Gozdzik et al. . Antibodies against F. hepatica were detected using either an in-house ELISA for liver fluke based on crude ES antigen following the method described by Salimi-Bejestani et al.  or the commercially available DRG liver fluke ELISAs (using recombinant Cathepsin L1). For the detection of C. oncophora antibodies no commercial ELISA is currently available and therefore no comparison was conducted.
GraphPad Prism® software 5.04 was used for the statistical analyses. Five parameter logistic regression curves were calculated in order to determine the optimal serum dilution for each bead set separately. For differentiation, positive and negative samples as well as sera from animals infected with non-target species were compared using box plots. Negative and positive cut-off MFI values for each parasite specific assay were obtained using receiver operating characteristics (ROC) analysis to determine at the same time sensitivity and specificity. The respective values are automatically provided with the 95 % confidence intervals in the software used.
For comparison with existing ELISA (F. hepatica and D. viviparus) two subsets of 39 and 370 serum samples, respectively, were examined and Kappa tests were performed.
Percentages with confidence intervals and differences between countries were calculated using a Mid-P exact test in OpenEpi (http://www.openepi.com/Menu/OE_Menu.htm).
Optimal amount of protein for coupling
The optimal amount of antigen identified for the target species were 0.45 μg for both, D. viviparus and C. oncophora, and 2.5 μg for F. hepatica. These were determined based on the amount of conjugated protein that would provide a reliable and reproducible MFI signal, which enabled a clear differentiation between positive and negative serum control samples.
Optimisation of secondary antibody dilutions and SA-PE concentration
Using the optimal amount of antigen, four different secondary antibody dilutions (1:500, 1:1000, 1:2000 and 1:5000) and four SA-PE concentrations (0.5 μg/ml, 1 μg/ml, 2 μg/ml and 4 μg/ml) were tested. Optimal results were obtained for a 1:1000 dilution for the biotinylated secondary antibody and a concentration of 2 μg/ml for SA-PE.
Optimal serum dilution
For all assays R2 values were close to 1. To achieve an optimal discrimination between positive and negative sera for all three bead sets a dilution of 1:200 was chosen.
Assessment of cross-reactivity and cut-off determination
Results of the Receiver Operating Characteristics analysis for negative (neg.) and positive (pos.) cut-off values. Median fluorescence intensity (MFI), sensitivity and specificity with confidence intervals (CI) are shown
95 % CI
95 % CI
Results of technical reproducibility using serum from different experimentally infected animals
Mean MFI value
Mean MFI value
Mean MFI value
Comparison between multiplex assay and single ELISAs and field validation
The validation of this multiplex assay was performed by comparing the results obtained from the examination of individual serum samples with already existing and established assays.
Initially, only positive and negative samples derived from experimentally infected animals pre and post infection (F. hepatica and D. viviparus) were compared, using an in-house ELISA for lungworm as described by von Holtum et al.  as well as an in-house ELISA for liver fluke based on crude ES antigen. The results obtained were identical for both, negative as well as positive control sera.
To increase the number of samples tested, additionally serum samples collected during a field trial in Poland were used for comparison. Thirty-nine of these samples were analysed using the commercially available DRG liver fluke ELISAs. For the detection of lungworm no commercial ELISA is currently available. Serum samples were analysed in parallel in Sweden using an in-house ELISA . In the latter, all samples were negative for D. viviparus antibodies while one sample was detected positive in the Luminex assay. Due to the fact that almost all samples were negative, no kappa statistic could be calculated. The comparison with the DRG liver fluke ELISA resulted in a kappa value of 0.37 (62.7 % of 0.60 maximum achievable). For this calculation the negative cut-off values for both assays were used. A larger subset of 363 samples were analysed using the commercially available SVANOVIR®F.hepatica-AbELISA (protocol identical to the in-house liver fluke ELISA as mentioned above) in the laboratory in Ghent. The comparison resulted in a kappa value of 0.460 (74.4 % of 0.67 maximal achievable). While 67 samples were positive only in the Luminex, there were also 21 samples which appeared positive in the ELISA but clearly negative in the Luminex assay.
Percentage of serum samples positive for antibodies against Cooperia oncophora, Dictyocaulus viviparus and Fasciola hepatica. Results for the field samples are shown per country and include 95 % confidence intervals (95 % CI)
Denmark (n = 39)
Poland (n = 367)
Switzerland (n = 76)
95 % CI
95 % CI
95 % CI
Pasture-borne parasitoses are highly prevalent in all grazing ruminants and have been recognised as important issues for animal welfare and productivity [2, 44]. This is particularly referring to the liver fluke, the lungworm and GI nematodes. Accelerated problems regarding anthelminthic resistance, climate change, intensification of farming and altered management practices have increased the need of development of new techniques in order to accurately diagnose and monitor these diseases.
In the current study, the successful development of a triplex assay for the simultaneous detection of antibodies against C. oncophora, D. viviparus and F. hepatica in serum from cattle is reported. While widely used singleplex assays using recombinant antigens were available for the latter two, expression of a C. oncophora antigen is described that was previously only once described for use in an ELISA. The new triplex assay was established using control sera from parasite naïve or controlled experimentally infected animals. Cross reactivity was assessed for all three antigens using the target parasites and revealed reliable detection with high sensitivity and specificity. Additionally, three non-target trichostrongylid nematode species (H. contortus, O. ostertagi and T. colubriformis) have been included in cross reactivity testing. Inclusion of Nematodirus, Paramphistomum and other important pasture-borne parasites would have been desirable, however, infection doses or serum of experimentally mono-infected animals were presently not available. Nevertheless, the evaluation of cross-reactivity presented here includes more parasite species than evaluated for many already commercialised assays.
Standardisation of any serological assay requires the use of defined negative and positive control sera collected from parasite naïve animals before and after specific mono-infection with the target parasites as well as non-target species. However, particularly regarding cross-reactivity, this information is often not accessible for commercially available assays. Also for the large number of developed and evaluated in-house ELISAs assessment of cross reactivity has often not been performed or at least not been reported. The specificity of the newly developed Luminex® assay was evaluated by testing the bead sets with sera obtained from animals infected with other important species for livestock (H. contortus, O. ostertagi and T. colubriformis). No cross-reactivity was obtained for all non-target species and all bead sets. For C. oncophora and F. hepatica also no cross-reactivity was detected for the other target species. For the detection of C. oncophora the 14.2 kDa recombinant antigen as reported by Poot et al.  was used. These authors reported no cross reactivity when serum samples from animals mono-infected with O. ostertagi or D. viviparus were used in their ELISA. This mirrors the result obtained in the current study, where no cross reactions were observed for any of the parasite species tested, leading to a sensitivity and specificity of 100 %.
For the detection of F. hepatica the recombinant Cathepsin L1 antigen was used. No cross reactivity was observed for any of the other species tested, also resulting a 100 % sensitivity and 100 % specificity. This is different to what has been reported in the literature regarding the use of this antigen. Kuerpick and colleagues  observed two false positive results out of 13 animals infected with D. viviparus and one out of four animals infected with C. oncophora (sensitivity between 90–100 %, specificity 88.6 %). Cornelissen et al.  obtained similar findings with five out of 191 animals infected with D. viviparus, one out of 31 animals infected with C. oncophora and one out of 55 animals infected with O. ostertagi (sensitivity 99.1 %, specificity 98.5 %). The currently observed absence of cross reactivity might not be confirmed in the Luminex® assay upon the use of significantly higher number of animals infected with non-target species. However, a similar level of cross reactivity as described above in the ELISAs would still be an improvement in comparison to systems using complex antigens such as ES antigen with reported specificities between 83–96 % .
The D. viviparus bead set was coupled with the recombinant MSP antigen. Although no or only minimal cross reactivity have been reported by von Holtum et al.  (sensitivity and specificity >99 %) as well as Gozdzik et al.  (sensitivity 97.7 %, specificity 98.1 %) for ELISAs using the same recombinant antigen, a few MFI values obtained for sera from animals infected with C. oncophora were very close to the lowest values of D. viviparus infected animals. However, no overlaps of MFI values occurred. In comparison to sera from negative animals or animals infected with O. ostertagi or H. contortus, MFI values obtained for sera from animals infected with F. hepatica and C. oncophora were substantially elevated. This background prevented the definition of two cut-off values separating clearly negative and clearly positive from the intermediate grey zone. The absence of two cut-off values for D. viviparus might complicate interpretation of field data in future epidemiological studies. In the field situation, some animals will be in the prepatency period or have already cured the infection but still have elevated antibody titres. MFI values for such animals can be expected to fall in the intermediate zone between negative and positive cut-off. Without such an intermediate zone, it will become difficult to identify such animals. Regarding D. viviparus it is also important to keep in mind that vaccination with attenuated larvae is possible. A cross reactivity of serum from vaccinated cattle is not expected since MSP is not expressed in premature stages and adult parasites do usually not develop. Additionally a sub-unit vaccine showing partial protection has been described recently . Here cross reactivity should also not occur since entirely different recombinant antigens are used. However, experimental evidence for this is not available yet.
Out of the currently available coproscopical, serological and molecular techniques for the diagnosis and the identification of parasites only the first two are routinely used. The sensitivity and specificity of the different techniques are extremely variable. Serological assays have the advantage of their implementation as herd health monitoring tools. All currently available assays are only capable of diagnosing antibodies against one parasite and are often limited in their specificity due to the use of complex crude or ES antigens. Despite the fact that for all three parasite species of interest, C. oncophora, D. viviparus and F. hepatica, ELISAs using recombinant antigens have been described, only for the latter is a commercial ELISA using recombinant antigen available. Multiplexing diagnostic systems are well established in human medicine, but in veterinary medicine this technology has so far not been extensively used. However, during recent years several commercial or “in-house” bead-based multiplex assays have been developed, particularly in the field of virology [39, 47, 48], where existing ELISAs were compared with the new multiplexing assays revealing generally better performance for the latter. Additionally, xMAP technology has been used for the detection of inflammatory markers, [49–52]. In the field of parasitology, serological multiplex assays have only been reported for the simultaneous detection of Trichinella spiralis and Toxoplasma gondii in pigs [40, 41] and for the detection of Plasmodium falciparum in humans . The Luminex® technology offers the advantage of reduced volume needed for diagnosis . When multi-plexing assays, the costs per analyte are considerably lower in the Luminex® assay than in a standard ELISA, whereas the time needed for the examination of the samples is generally similar but reduced labour time per assay is achieved.
After the establishment of the Luminex assay, field samples obtained from naturally infected grazing animals during a trial in Poland have been used for comparison of the new Luminex® assay with already established ELISAs. In the case of multiplex bead-based immunoassays, comparison of the Luminex is frequently made with serological assays, such as ELISA. Correlation of these two methods varies among literature [34, 54–59]. In the current study, agreement between the assays was generally good when the same (recombinant) antigen was used. In case of the lungworm assay it was almost impossible to calculate kappa values since only one animal out of the 39 samples tested appeared positive (only in the new assay). The comparison of the DRG ELISA with Luminex was complicated by the fact that this ELISA uses five different categories and it was unclear, how to handle samples negative or positive in the Luminex assay and “questionable” or “low to moderate infection” in the other. Most of the deviations in classification were due to the fact, that the number of positive samples was higher in the Luminex assay but there were also one sample classified as negative in the Luminex but positive with the ELISA. Comparison of the Luminex® data with the ELISA using the ES antigen for F. hepatica detection had the advantage of a high number of samples tested. This resulted in a better kappa value than that obtained for the first comparison, however, there was a high number of samples positive in only one of the assays while being clearly negative (ES ELISA) or in the grey zone (Luminex®) in the respective other one. Since a recombinant antigen was used in the Luminex® assay a higher specificity can be expected than using a complex ES antigen. At the same time, this might result in decreased sensitivity since not all animals might develop antibodies against Cathepsin L1. However, more samples were detected positive with the Luminex® assay (67 vs 21), so sensitivity should not be of great concern here. Another explanation for the data obtained might be false positive samples in the ES ELISA, for example due to infection with closely related pathogens.
A multiplex fluorescence serum immunoassay was successfully developed for the simultaneous detection of antibodies produced against C. oncophora, D. viviparus and F. hepatica in serum samples from cattle. It is characterised by low cost and time, high reproducibility and more importantly, by high sensitivity and specificity due to the use of recombinant antigens. This platform provides rapid, high-throughput diagnostic results, allowing the incorporation of further analytes and parameters in the future, such as hormones, inflammatory biomarkers and additional parasites. In particular, inclusion of O. ostertagi should be a major aim for the future, though recombinant antigens suitable for diagnosis are currently not available.
The financial support from the EU funded FP 7 project GLOWORM (KBBE 2011.1.3-04, No 288975) is gratefully acknowledged.
- Morgan E, Charlier J, Hendrickx G, Biggeri A, Catalan D, von Samson-Himmelstjerna G, et al. Global Change and Helminth Infections in Grazing Ruminants in Europe: Impacts, Trends and Sustainable Solutions. Agriculture. 2013;3(3):484–502.View ArticleGoogle Scholar
- Charlier J, Hoglund J, von Samson-Himmelstjerna G, Dorny P, Vercruysse J. Gastrointestinal nematode infections in adult dairy cattle: impact on production, diagnosis and control. Vet Parasitol. 2009;164(1):70–9.PubMedView ArticleGoogle Scholar
- Alvarez Rojas CA, Jex AR, Gasser RB, Scheerlinck JP. Techniques for the diagnosis of Fasciola infections in animals: room for improvement. Adv Parasitol. 2014;85:65–107.PubMedView ArticleGoogle Scholar
- Barda BD, Rinaldi L, Ianniello D, Zepherine H, Salvo F, Sadutshang T, et al. Mini-FLOTAC, an innovative direct diagnostic technique for intestinal parasitic infections: experience from the field. PLoS Negl Trop Dis. 2013;7(8):e2344.PubMed CentralPubMedView ArticleGoogle Scholar
- Bisset SA. Helminth parasites of economic importance in cattle in New Zealand. N Z J Zool. 1994;21(1):9–22.View ArticleGoogle Scholar
- Piekarska J, Ploneczka-Janeczko K, Kantyka M, Kuczaj M, Gorczykowski M, Janeczko K. Gastrointestinal nematodes in grazing dairy cattle from small and medium-sized farms in southern Poland. Vet Parasitol. 2013;198(1–2):250–3.PubMedView ArticleGoogle Scholar
- Coles GC. Cattle nematodes resistant to anthelmintics: why so few cases? Vet Res. 2002;33(5):481–9.PubMedView ArticleGoogle Scholar
- Dimander SO, Hoglund J, Sporndly E, Waller PJ. The impact of internal parasites on the productivity of young cattle organically reared on semi-natural pastures in Sweden. Vet Parasitol. 2000;90(4):271–84.PubMedView ArticleGoogle Scholar
- Gibb MJ, Huckle CA, Forbes AB. Effects of sequential treatments with eprinomectin on performance and grazing behaviour in dairy cattle under daily-paddock stocking management. Vet Parasitol. 2005;133(1):79–90.PubMedView ArticleGoogle Scholar
- Sanchez J, Dohoo I, Carrier J, DesCoteaux L. A meta-analysis of the milk-production response after anthelmintic treatment in naturally infected adult dairy cows. Prev Vet Med. 2004;63(3–4):237–56.PubMedView ArticleGoogle Scholar
- Schunn A-M, Conraths FJ, Staubach C, Fröhlich A, Forbes A, Schnieder T, et al. Lungworm Infections in German Dairy Cattle Herds — Seroprevalence and GIS-Supported Risk Factor Analysis. PLoS ONE. 2013;8(9):e74429.PubMed CentralPubMedView ArticleGoogle Scholar
- Perry BD, Randolph TF. Improving the assessment of the economic impact of parasitic diseases and of their control in production animals. Vet Parasitol. 1999;84(3–4):145–68.PubMedView ArticleGoogle Scholar
- Nieuwhof GJ, Bishop SC. Costs of the major endemic diseases of sheep in Great Britain and the potential benefits of reduction in disease impact. Anim Sci. 2005;81(01):23–9.View ArticleGoogle Scholar
- Holzhauer M, van Schaik G, Saatkamp HW, Ploeger HW. Lungworm outbreaks in adult dairy cows: estimating economic losses and lessons to be learned. Vet Rec. 2011;169(19):494.PubMedView ArticleGoogle Scholar
- Schweizer G, Braun U, Deplazes P, Torgerson PR. The economic effects of bovine fasciolosis in Switzerland. Vet Rec. 2005;157(7):188–93.PubMedView ArticleGoogle Scholar
- Spithill TW, Smooker PM, Sexton JL, Bozas E, Morrison CA, Creany J, et al. Development of vaccines against Fasciola hepatica. In: Dalton JP, editor. Fasciolosis. Wallingford, UK: CAB International Publishing; 1999. p. 377–410.Google Scholar
- Fürst T, Keiser J, Utzinger J. Global burden of human food-borne trematodiasis: a systematic review and meta-analysis. Lancet Infect Dis. 2012;12(3):210–21.PubMedView ArticleGoogle Scholar
- Bott NJ, Campbell BE, Beveridge I, Chilton NB, Rees D, Hunt PW, et al. A combined microscopic-molecular method for the diagnosis of strongylid infections in sheep. Int J Parasitol. 2009;39(11):1277–87.PubMedView ArticleGoogle Scholar
- Demeler J, Ramunke S, Wolken S, Ianiello D, Rinaldi L, Gahutu JB, et al. Discrimination of gastrointestinal nematode eggs from crude fecal egg preparations by inhibitor-resistant conventional and real-time PCR. PLoS One. 2013;8(4):e61285.PubMed CentralPubMedView ArticleGoogle Scholar
- Roeber F, Jex AR, Campbell AJ, Campbell BE, Anderson GA, Gasser RB. Evaluation and application of a molecular method to assess the composition of strongylid nematode populations in sheep with naturally acquired infections. Infect Genet Evol. 2011;11(5):849–54.PubMedView ArticleGoogle Scholar
- Charlier J, De Meulemeester L, Claerebout E, Williams D, Vercruysse J. Qualitative and quantitative evaluation of coprological and serological techniques for the diagnosis of fasciolosis in cattle. Vet Parasitol. 2008;153(1–2):44–51.PubMedView ArticleGoogle Scholar
- Kuerpick B, Schnieder T, Strube C. Evaluation of a recombinant cathepsin L1 ELISA and comparison with the Pourquier and ES ELISA for the detection of antibodies against Fasciola hepatica. Vet Parasitol. 2013;193(1–3):206–13.PubMedView ArticleGoogle Scholar
- Salimi-Bejestani MR, McGarry JW, Felstead S, Ortiz P, Akca A, Williams DJ. Development of an antibody-detection ELISA for Fasciola hepatica and its evaluation against a commercially available test. Res Vet Sci. 2005;78(2):177–81.PubMedView ArticleGoogle Scholar
- Boon JH, Ploeger HW, Raaymakers AJ. Sero-epidemiological survey of Dictyocaulus viviparus infections in first-season grazing calves in The Netherlands. Vet Rec. 1986;119(19):475–9.PubMedView ArticleGoogle Scholar
- Bos HJ, Beekman J. Serodiagnosis of lungworm infection in calves using ELISA. Dev Biol Stand. 1985;62:45–52.PubMedGoogle Scholar
- Marius V, Bernard S, Raynaud JP, Pery P, Luffau G. Dictyocaulus viviparus in calves: quantitation of antibody activities in sera and respiratory secretions by immuno-enzymatic analysis. Ann Rech Vet. 1979;10(1):55–63.PubMedGoogle Scholar
- Gozdzik K, Engstrom A, Hoglund J. Optimization of in-house ELISA based on recombinant major sperm protein (rMSP) of Dictyocaulus viviparus for the detection of lungworm infection in cattle. Res Vet Sci. 2012;93(2):813–8.PubMedView ArticleGoogle Scholar
- Schnieder T. Use of a recombinant Dictyocaulus viviparus antigen in an enzyme-linked immunosorbent assay for immunodiagnosis of bovine dictyocaulosis. Parasitol Res. 1992;78(4):298–302.PubMedView ArticleGoogle Scholar
- von Holtum C, Strube C, Schnieder T, von Samson-Himmelstjerna G. Development and evaluation of a recombinant antigen-based ELISA for serodiagnosis of cattle lungworm. Vet Parasitol. 2008;151(2–4):218–26.View ArticleGoogle Scholar
- Ploeger HW, Kloosterman A, Rietveld FW, Berghen P, Hilderson H, Hollanders W. Quantitative estimation of the level of exposure to gastrointestinal nematode infection in first-year calves. Vet Parasitol. 1994;55(4):287–315.PubMedView ArticleGoogle Scholar
- Poot J, Kooyman FN, Dop PY, Schallig HD, Eysker M, Cornelissen AW. Use of cloned excretory/secretory low-molecular-weight proteins of Cooperia oncophora in a serological assay. J Clin Microbiol. 1997;35(7):1728–33.PubMed CentralPubMedGoogle Scholar
- Fulton RJ, McDade RL, Smith PL, Kienker LJ, Kettman Jr JR. Advanced multiplexed analysis with the FlowMetrixTM system. Clin Chem. 1997;43(9):1749–56.PubMedGoogle Scholar
- Carson RT, Vignali DAA. Simultaneous quantitation of 15 cytokines using a multiplexed flow cytometric assay. J Immunol Methods. 1999;227(1–2):41–52.PubMedView ArticleGoogle Scholar
- Baker HN, Murphy R, Lopez E, Garcia C. Conversion of a capture ELISA to a Luminex xMAP assay using a multiplex antibody screening method. J Vis Exp. 2012;6:65.Google Scholar
- Kettman JR, Davies T, Chandler D, Oliver KG, Fulton RJ. Classification and properties of 64 multiplexed microsphere sets. Cytometry. 1998;33(2):234–43.PubMedView ArticleGoogle Scholar
- Earley MC, Vogt RF, Shapiro HM, Mandy FF, Kellar KL, Bellisario R, et al. Report from a workshop on multianalyte microsphere assays. Cytometry. 2002;50(5):239–42.PubMedView ArticleGoogle Scholar
- Dunbar SA. Applications of Luminex xMAP technology for rapid, high-throughput multiplexed nucleic acid detection. Clin Chim Acta. 2006;363(1–2):71–82.PubMedView ArticleGoogle Scholar
- Tighe PJ, Ryder RR, Todd I, Fairclough LC. ELISA in the Multiplex Era; Potential and Pitfalls. Proteomics Clin Appl. 2015;9(3-4):406–22.PubMedView ArticleGoogle Scholar
- Christopher-Hennings J, Araujo KP, Souza CJ, Fang Y, Lawson S, Nelson EA, et al. Opportunities for bead-based multiplex assays in veterinary diagnostic laboratories. J Vet Diagn Invest. 2013;25(6):671–91.PubMedView ArticleGoogle Scholar
- Bokken GC, Bergwerff AA, van Knapen F. A novel bead-based assay to detect specific antibody responses against Toxoplasma gondii and Trichinella spiralis simultaneously in sera of experimentally infected swine. BMC Vet Res. 2012;8:36.PubMed CentralPubMedView ArticleGoogle Scholar
- van der Wal FJ, Achterberg RP, Kant A, Maassen CB. A bead-based suspension array for the serological detection of Trichinella in pigs. Vet J. 2013;196(3):439–44.PubMedView ArticleGoogle Scholar
- Ducheyne E, Charlier J, Vercruysse J, Rinaldi L, Biggeri A, Demeler J, et al. Modelling the spatial distribution of Fasciola hepatica in dairy cattle in Europe. Geospat Health. 2015;9(2):261–70.PubMedView ArticleGoogle Scholar
- Collins PR, Stack CM, O’Neill SM, Doyle S, Ryan T, Brennan GP, et al. Cathepsin L1, the Major Protease Involved in Liver Fluke (Fasciola hepatica) Virulence: propeptide cleavage sites and autoactivation of the zymogen secreted from astrodermal cells. J Biol Chem. 2004;279(17):17038–46.PubMedView ArticleGoogle Scholar
- Charlier J, Vercruysse J, Morgan E, van Dijk J, Williams DJ. Recent advances in the diagnosis, impact on production and prediction of Fasciola hepatica in cattle. Parasitology. 2014;141(3):326–35.PubMedView ArticleGoogle Scholar
- Cornelissen JB, Borgsteede FH, van Milligen FJ. Evaluation of an ELISA for the routine diagnosis of Dictyocaulus viviparus infections in cattle. Vet Parasitol. 1997;70(1–3):153–64.PubMedView ArticleGoogle Scholar
- Strube C, Haake C, Sager H, Schorderet Weber S, Kaminsky R, Buschbaum S, et al. Vaccination with recombinant paramyosin against the bovine lungworm Dictyocaulus viviparus considerably reduces worm burden and larvae shedding. Parasit Vector. 2015;8:119.View ArticleGoogle Scholar
- Go YY, Wong SJ, Branscum AJ, Demarest VL, Shuck KM, Vickers ML, et al. Development of a fluorescent-microsphere immunoassay for detection of antibodies specific to equine arteritis virus and comparison with the virus neutralization test. Clin Vaccine Immunol. 2008;15(1):76–87.PubMed CentralPubMedView ArticleGoogle Scholar
- Langenhorst RJ, Lawson S, Kittawornrat A, Zimmerman JJ, Sun Z, Li Y, et al. Development of a fluorescent microsphere immunoassay for detection of antibodies against porcine reproductive and respiratory syndrome virus using oral fluid samples as an alternative to serum-based assays. Clin Vaccine Immunol. 2012;19(2):180–9.PubMed CentralPubMedView ArticleGoogle Scholar
- Johannisson A, Jonasson R, Dernfalk J, Jensen-Waern M. Simultaneous detection of porcine proinflammatory cytokines using multiplex flow cytometry by the xMAP technology. Cytometry A. 2006;69(5):391–5.PubMedView ArticleGoogle Scholar
- Bjerre M, Hansen TK, Flyvbjerg A, Tonnesen E. Simultaneous detection of porcine cytokines by multiplex analysis: development of magnetic bioplex assay. Vet Immunol Immunopathol. 2009;130(1–2):53–8.PubMedView ArticleGoogle Scholar
- Wyns H, Croubels S, Demeyere K, Watteyn A, De Backer P, Meyer E. Development of a cytometric bead array screening tool for the simultaneous detection of pro-inflammatory cytokines in porcine plasma. Vet Immunol Immunopathol. 2013;151(1–2):28–36.PubMedView ArticleGoogle Scholar
- Bongoni AK, Lanz J, Rieben R, Banz Y. Development of a bead-based multiplex assay for the simultaneous detection of porcine inflammation markers using xMAP technology. Cytometry A. 2013;83(7):636–47.PubMedView ArticleGoogle Scholar
- Perraut R, Richard V, Varela ML, Trape JF, Guillotte M, Tall A, et al. Comparative analysis of IgG responses to Plasmodium falciparum MSP1p19 and PF13-DBL1alpha1 using ELISA and a magnetic bead-based duplex assay (MAGPIX(R)-Luminex) in a Senegalese meso-endemic community. Malar J. 2014;13:410.PubMed CentralPubMedView ArticleGoogle Scholar
- Klein DL, Martinez JE, Hickey MH, Hassouna F, Zaman K, Steinhoff M. Development and characterization of a multiplex bead-based immunoassay to quantify pneumococcal capsular polysaccharide-specific antibodies. Clin Vaccine Immunol. 2012;19(8):1276–82.PubMed CentralPubMedView ArticleGoogle Scholar
- Dernfalk J, Persson Waller K, Johannisson A. The xMAP technique can be used for detection of the inflammatory cytokines IL-1beta, IL-6 and TNF-alpha in bovine samples. Vet Immunol Immunopathol. 2007;118(1–2):40–9.PubMedView ArticleGoogle Scholar
- Elshal MF, McCoy JP. Multiplex bead array assays: performance evaluation and comparison of sensitivity to ELISA. Methods. 2006;38(4):317–23.PubMed CentralPubMedView ArticleGoogle Scholar
- Lal G, Balmer P, Stanford E, Martin S, Warrington R, Borrow R. Development and validation of a nonaplex assay for the simultaneous quantitation of antibodies to nine Streptococcus pneumoniae serotypes. J Immunol Methods. 2005;296(1–2):135–47.PubMedView ArticleGoogle Scholar
- Pickering JW, Martins TB, Schroder MC, Hill HR. Comparison of a Multiplex Flow Cytometric Assay with Enzyme-Linked Immunosorbent Assay for Quantitation of Antibodies to Tetanus, Diphtheria, and Haemophilus influenzae Type b. Clin Diagn Lab Immunol. 2002;9(4):872–6.PubMed CentralPubMedGoogle Scholar
- Willman JH, Hill HR, Martins TB, Jaskowski TD, Ashwood ER, Litwin CM. Multiplex analysis of heterophil antibodies in patients with indeterminate HIV immunoassay results. Am J Clin Pathol. 2001;115(5):764–9.PubMedView ArticleGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.